TEST#,REC#,DATE/TIME,REMOTE#,SOCKET#,FORM,PHASE,TA,Kh,MODE,SENSOR,METER ID:,%HL,%LL,%WT,%PF,CREEP,Question1,Answer1,Question2,Answer2,Question3,Answer3,Question4,Answer4,Question5,Answer5,Question6,Answer6,Question7,Answer7,Question8,Answer8,RevHL,RevLL,RevPF,RatioWT,AmpHL,AmpLL,AmpPF,WhHL,WhLL,WhPF,VOLT,FREQ,LATITUDE,LONGITUDE,ALTITUDE
180,180,9/5/2019 14:27,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40451767,99.94%,99.97%,99.95%,99.92%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.04 Wh,12.00 Wh,36.03 Wh,245.51 Vrms,60.03 Hz,0,0,0
179,179,9/5/2019 13:05,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40585229,100.01%,100.02%,100.01%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.00 Wh,35.99 Wh,246.92 Vrms,60.01 Hz,0,0,0
178,178,9/5/2019 11:55,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40283556,99.99%,100.01%,99.99%,100.00%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.01 Wh,12.00 Wh,36.00 Wh,245.06 Vrms,60.02 Hz,0,0,0
177,177,9/5/2019 11:37,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442519,99.99%,100.00%,99.99%,99.87%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.01 Wh,12.00 Wh,36.05 Wh,244.96 Vrms,60.00 Hz,0,0,0
176,176,9/5/2019 10:58,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40313452,99.92%,99.92%,99.92%,99.85%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.06 Wh,12.01 Wh,36.05 Wh,246.10 Vrms,60.02 Hz,0,0,0
175,175,9/5/2019 10:39,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40585303,99.95%,99.98%,99.96%,99.93%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.03 Wh,12.00 Wh,36.02 Wh,244.71 Vrms,59.98 Hz,0,0,0
174,174,8/5/2019 15:14,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40639911,99.99%,100.04%,100.00%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.21 Wh,7.20 Wh,21.60 Wh,244.28 Vrms,60.02 Hz,0,0,0
173,173,8/5/2019 15:01,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40564498,100.04%,100.07%,100.05%,100.16%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.19 Wh,21.57 Wh,244.61 Vrms,60.02 Hz,0,0,0
172,172,8/5/2019 14:46,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40565549,100.01%,100.02%,100.01%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.59 Wh,244.13 Vrms,59.99 Hz,0,0,0
171,171,8/5/2019 14:29,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40568090,99.99%,100.00%,99.99%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.60 Wh,245.41 Vrms,60.00 Hz,0,0,0
170,170,8/5/2019 14:05,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40643829,99.99%,100.01%,100.00%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.60 Wh,244.70 Vrms,60.02 Hz,0,0,0
169,169,8/5/2019 13:36,12538318,12538318,16S,ABC,50.00 A,21.6,quick,Optical Pickup,40470432,99.88%,99.87%,99.87%,99.84%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,9 pulses,1 pulses,5 pulses,4,50.00 A,5.00 A,50.00 A,194.63 Wh,21.63 Wh,108.17 Wh,211.68 Vrms,60.00 Hz,0,0,0
168,168,8/5/2019 13:20,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40634388,100.02%,100.05%,100.03%,99.95%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.61 Wh,245.33 Vrms,60.00 Hz,0,0,0
167,167,6/5/2019 16:00,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40624989,100.05%,100.06%,100.05%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.20 Wh,21.59 Wh,244.93 Vrms,60.03 Hz,0,0,0
166,166,6/5/2019 15:40,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40627138,100.10%,100.11%,100.10%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.16 Wh,7.19 Wh,21.59 Wh,246.21 Vrms,59.99 Hz,0,0,0
165,165,6/5/2019 15:20,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40746185,100.10%,100.11%,100.10%,100.09%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.93 Wh,11.99 Wh,35.97 Wh,244.98 Vrms,60.01 Hz,0,0,0
164,164,6/5/2019 11:41,12538318,12538318,16S,ABC,50.00 A,21.6,quick,Optical Pickup,40441522,100.18%,100.20%,100.18%,100.17%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,10 pulses,1 pulses,5 pulses,4,50.00 A,5.00 A,50.00 A,215.62 Wh,21.56 Wh,107.82 Wh,237.55 Vrms,60.01 Hz,0,0,0
163,163,6/5/2019 11:24,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40440046,100.00%,100.01%,100.01%,100.04%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,129.60 Wh,21.60 Wh,64.78 Wh,245.32 Vrms,59.98 Hz,0,0,0
162,162,6/5/2019 10:12,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40396131,100.00%,100.03%,100.01%,99.95%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.61 Wh,247.66 Vrms,60.00 Hz,0,0,0
161,161,3/5/2019 15:09,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40562522,100.01%,100.04%,100.02%,99.94%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.61 Wh,243.44 Vrms,60.02 Hz,0,0,0
160,160,3/5/2019 13:57,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40625824,99.99%,100.01%,99.99%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.21 Wh,7.20 Wh,21.59 Wh,243.69 Vrms,59.99 Hz,0,0,0
159,159,3/5/2019 13:38,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40634080,100.03%,100.02%,100.03%,100.00%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.60 Wh,244.42 Vrms,60.02 Hz,0,0,0
158,158,3/5/2019 12:41,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40452652,100.03%,100.00%,100.02%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.60 Wh,239.86 Vrms,59.99 Hz,0,0,0
157,157,2/5/2019 14:26,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40576853,99.99%,99.99%,99.99%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.60 Wh,248.85 Vrms,60.00 Hz,0,0,0
156,156,2/5/2019 14:07,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40559949,100.10%,100.13%,100.11%,100.15%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.16 Wh,7.19 Wh,21.57 Wh,250.18 Vrms,60.01 Hz,0,0,0
155,155,2/5/2019 13:49,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40559345,99.97%,99.98%,99.97%,99.95%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.21 Wh,7.20 Wh,21.61 Wh,247.84 Vrms,60.01 Hz,0,0,0
154,154,2/5/2019 13:21,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40549284,100.07%,100.08%,100.07%,100.10%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.17 Wh,7.19 Wh,21.58 Wh,247.91 Vrms,59.97 Hz,0,0,0
153,153,2/5/2019 13:01,12538318,12538318,1S,AC,15.00 A,1.8,quick,Optical Pickup,40487037,100.03%,99.99%,100.02%,100.10%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,15.00 A,1.50 A,15.00 A,10.80 Wh,1.80 Wh,5.39 Wh,124.00 Vrms,60.02 Hz,0,0,0
152,152,22/04/2019 14:11,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40439530,99.99%,100.03%,100.01%,100.04%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,108.01 Wh,21.59 Wh,64.78 Wh,212.82 Vrms,60.01 Hz,0,0,0
151,151,22/04/2019 13:33,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40452673,100.03%,100.01%,100.03%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.59 Wh,244.96 Vrms,59.98 Hz,0,0,0
150,150,22/04/2019 11:27,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40635379,100.06%,100.07%,100.06%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.17 Wh,7.19 Wh,21.60 Wh,236.92 Vrms,60.00 Hz,0,0,0
149,149,22/04/2019 11:07,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40486914,100.08%,100.04%,100.08%,100.10%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.16 Wh,7.20 Wh,21.58 Wh,245.41 Vrms,60.00 Hz,0,0,0
148,148,22/04/2019 10:30,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40634678,100.04%,100.02%,100.04%,100.08%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.20 Wh,21.58 Wh,238.92 Vrms,60.02 Hz,0,0,0
147,147,19/04/2019 12:48,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40636410,100.08%,100.09%,100.08%,100.12%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.17 Wh,7.19 Wh,21.57 Wh,246.74 Vrms,60.01 Hz,0,0,0
146,146,19/04/2019 12:28,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40639061,99.97%,99.98%,99.97%,99.93%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.21 Wh,7.20 Wh,21.62 Wh,246.04 Vrms,59.99 Hz,0,0,0
145,145,19/04/2019 12:04,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40477333,100.03%,100.06%,100.03%,100.05%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.59 Wh,246.51 Vrms,60.00 Hz,0,0,0
144,144,19/04/2019 11:44,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40605383,99.86%,99.85%,99.86%,99.81%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.26 Wh,7.21 Wh,21.64 Wh,246.57 Vrms,60.00 Hz,0,0,0
143,143,19/04/2019 11:25,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40631469,99.98%,100.00%,99.99%,99.94%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.21 Wh,7.20 Wh,21.61 Wh,246.64 Vrms,59.99 Hz,0,0,0
142,142,19/04/2019 11:05,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40562760,100.03%,100.04%,100.04%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,23.91 A,3.00 A,23.91 A,35.99 Wh,7.20 Wh,21.59 Wh,248.65 Vrms,59.99 Hz,0,0,0
141,141,19/04/2019 10:38,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40562674,100.00%,100.04%,100.01%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.60 Wh,249.63 Vrms,60.01 Hz,0,0,0
140,140,19/04/2019 10:19,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40452726,100.00%,99.99%,100.00%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.60 Wh,247.67 Vrms,59.96 Hz,0,0,0
139,139,18/04/2019 12:40,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40471171,100.11%,100.13%,100.11%,100.16%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.92 Wh,11.98 Wh,35.94 Wh,245.00 Vrms,59.98 Hz,0,0,0
138,138,18/04/2019 12:10,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40454846,100.04%,100.00%,100.03%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.20 Wh,21.59 Wh,247.66 Vrms,60.02 Hz,0,0,0
137,137,18/04/2019 11:46,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40585318,100.02%,100.03%,100.02%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.00 Wh,36.00 Wh,244.99 Vrms,59.98 Hz,0,0,0
136,136,18/04/2019 11:20,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40602228,99.81%,99.84%,99.81%,99.79%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.28 Wh,7.21 Wh,21.64 Wh,248.32 Vrms,60.01 Hz,0,0,0
135,135,18/04/2019 10:44,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40494150,100.02%,100.01%,100.02%,99.97%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.00 Wh,36.01 Wh,244.39 Vrms,59.99 Hz,0,0,0
134,134,18/04/2019 10:20,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40471080,99.96%,99.96%,99.96%,99.95%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.03 Wh,12.00 Wh,36.02 Wh,244.42 Vrms,59.98 Hz,0,0,0
133,133,18/04/2019 10:00,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40585314,100.01%,100.02%,100.01%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.00 Wh,12.00 Wh,35.99 Wh,243.36 Vrms,59.98 Hz,0,0,0
132,132,17/04/2019 16:10,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40377820,100.04%,100.05%,100.04%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,107.96 Wh,21.59 Wh,64.76 Wh,210.41 Vrms,60.00 Hz,0,0,0
131,131,17/04/2019 15:48,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40485769,100.02%,100.01%,100.01%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.00 Wh,36.01 Wh,244.26 Vrms,59.99 Hz,0,0,0
130,130,17/04/2019 15:29,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40746188,100.07%,100.08%,100.07%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.95 Wh,11.99 Wh,35.98 Wh,244.69 Vrms,60.02 Hz,0,0,0
129,129,16/04/2019 14:29,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40451747,100.14%,100.17%,100.14%,100.13%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.90 Wh,11.98 Wh,35.95 Wh,246.59 Vrms,60.02 Hz,0,0,0
128,128,16/04/2019 14:10,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40625631,100.04%,100.05%,100.04%,99.93%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.20 Wh,21.61 Wh,243.02 Vrms,60.00 Hz,0,0,0
127,127,16/04/2019 13:52,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40585271,100.11%,100.14%,100.12%,100.10%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.92 Wh,11.98 Wh,35.96 Wh,243.93 Vrms,60.00 Hz,0,0,0
126,126,16/04/2019 12:19,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40494087,100.06%,100.07%,100.06%,100.04%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.96 Wh,11.99 Wh,35.98 Wh,248.01 Vrms,60.01 Hz,0,0,0
125,125,16/04/2019 11:41,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40433619,100.18%,100.19%,100.18%,100.20%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.87 Wh,11.98 Wh,35.93 Wh,246.73 Vrms,60.01 Hz,0,0,0
124,124,16/04/2019 11:14,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40313592,100.08%,100.09%,100.09%,100.05%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.94 Wh,11.99 Wh,35.98 Wh,251.47 Vrms,60.03 Hz,0,0,0
123,123,15/04/2019 12:31,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40471138,100.01%,100.01%,100.01%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.00 Wh,36.01 Wh,252.31 Vrms,59.97 Hz,0,0,0
122,122,15/04/2019 12:12,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40441406,100.01%,100.02%,100.02%,99.99%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.00 Wh,36.00 Wh,252.33 Vrms,60.01 Hz,0,0,0
121,121,15/04/2019 11:54,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40313503,100.15%,100.19%,100.16%,100.12%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.89 Wh,11.98 Wh,35.96 Wh,251.44 Vrms,60.02 Hz,0,0,0
120,120,15/04/2019 11:39,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442566,100.05%,100.07%,100.06%,100.10%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.96 Wh,11.99 Wh,35.97 Wh,251.43 Vrms,59.98 Hz,0,0,0
119,119,15/04/2019 11:23,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442567,100.04%,100.04%,100.04%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.97 Wh,12.00 Wh,35.99 Wh,248.57 Vrms,60.00 Hz,0,0,0
118,118,11/4/2019 13:27,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40313541,100.02%,100.04%,100.02%,100.05%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,11.99 Wh,35.98 Wh,240.55 Vrms,60.03 Hz,0,0,0
117,117,11/4/2019 13:09,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40471308,100.04%,100.05%,100.04%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.97 Wh,11.99 Wh,35.99 Wh,243.22 Vrms,60.02 Hz,0,0,0
116,116,11/4/2019 12:50,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442562,100.03%,100.06%,100.04%,100.11%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.98 Wh,11.99 Wh,35.96 Wh,243.86 Vrms,60.02 Hz,0,0,0
115,115,11/4/2019 12:30,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442568,100.33%,100.35%,100.34%,100.29%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.76 Wh,11.96 Wh,35.89 Wh,243.18 Vrms,59.99 Hz,0,0,0
114,114,11/4/2019 12:06,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40585300,100.04%,100.06%,100.04%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.97 Wh,11.99 Wh,35.99 Wh,246.86 Vrms,60.02 Hz,0,0,0
113,113,11/4/2019 11:33,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40544252,100.01%,99.95%,100.00%,99.94%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.99 Wh,12.01 Wh,36.02 Wh,245.95 Vrms,60.02 Hz,0,0,0
112,112,11/4/2019 11:09,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442512,100.12%,100.12%,100.12%,100.09%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,71.92 Wh,11.99 Wh,35.97 Wh,245.56 Vrms,59.99 Hz,0,0,0
111,111,11/4/2019 10:50,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442513,100.00%,100.00%,100.00%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.00 Wh,12.00 Wh,36.01 Wh,247.18 Vrms,60.01 Hz,0,0,0
110,110,11/4/2019 10:04,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40442591,99.97%,99.99%,99.97%,99.97%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.02 Wh,12.00 Wh,36.01 Wh,249.21 Vrms,59.98 Hz,0,0,0
109,109,10/4/2019 14:48,12538318,12538318,16S,ABC,50.00 A,21.6,quick,Optical Pickup,40442769,99.94%,99.93%,99.94%,99.97%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,9 pulses,1 pulses,5 pulses,4,50.00 A,5.00 A,50.00 A,194.52 Wh,21.61 Wh,108.04 Wh,214.78 Vrms,60.00 Hz,0,0,0
108,108,10/4/2019 12:53,12538318,12538318,16S,ABC,50.00 A,21.6,quick,Optical Pickup,40442768,100.11%,100.11%,100.10%,100.05%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,9 pulses,1 pulses,5 pulses,4,50.00 A,5.00 A,50.00 A,194.18 Wh,21.58 Wh,107.95 Wh,215.34 Vrms,60.02 Hz,0,0,0
107,107,10/4/2019 12:35,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40469920,100.07%,100.08%,100.08%,100.11%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,107.92 Wh,21.58 Wh,64.73 Wh,212.51 Vrms,60.00 Hz,0,0,0
106,106,10/4/2019 12:23,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40469922,100.03%,100.00%,100.02%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,107.97 Wh,21.60 Wh,64.79 Wh,212.78 Vrms,60.00 Hz,0,0,0
105,105,10/4/2019 12:07,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40250149,99.98%,100.03%,99.99%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.21 Wh,7.20 Wh,21.60 Wh,245.40 Vrms,60.02 Hz,0,0,0
104,104,10/4/2019 11:49,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40399653,100.02%,100.04%,100.02%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.59 Wh,245.80 Vrms,59.98 Hz,0,0,0
103,103,10/4/2019 11:30,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40399658,100.00%,100.02%,100.00%,99.94%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.61 Wh,245.85 Vrms,60.01 Hz,0,0,0
102,102,10/4/2019 10:34,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40400312,100.00%,100.02%,100.00%,99.99%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.20 Wh,7.20 Wh,21.60 Wh,239.96 Vrms,59.87 Hz,0,0,0
101,101,10/4/2019 10:07,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40400310,100.02%,100.04%,100.03%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.19 Wh,7.20 Wh,21.60 Wh,242.16 Vrms,59.99 Hz,0,0,0
100,100,9/4/2019 16:09,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40644807,100.07%,100.07%,100.07%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.17 Wh,7.20 Wh,21.59 Wh,244.18 Vrms,59.98 Hz,0,0,0
99,99,9/4/2019 15:29,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,22324289,100.09%,100.10%,100.10%,100.07%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.16 Wh,7.19 Wh,21.59 Wh,240.74 Vrms,60.00 Hz,0,0,0
98,98,9/4/2019 12:45,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40602026,100.05%,100.06%,100.05%,100.12%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.20 Wh,21.57 Wh,250.50 Vrms,59.97 Hz,0,0,0
97,97,9/4/2019 12:20,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40602023,100.08%,100.12%,100.09%,100.07%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.17 Wh,7.19 Wh,21.58 Wh,250.07 Vrms,59.98 Hz,0,0,0
96,96,9/4/2019 12:00,12538318,12538318,16S,ABC,50.00 A,21.6,quick,Optical Pickup,40435870,99.99%,99.96%,99.98%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,20 pulses,2 pulses,10 pulses,4,50.00 A,5.00 A,50.00 A,432.04 Wh,43.22 Wh,216.03 Wh,496.54 Vrms,60.02 Hz,0,0,0
95,95,9/4/2019 11:31,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40561386,100.08%,100.07%,100.08%,100.10%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.16 Wh,7.19 Wh,21.58 Wh,245.65 Vrms,60.02 Hz,0,0,0
94,94,8/4/2019 13:47,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40377621,100.03%,100.04%,100.04%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,107.96 Wh,21.59 Wh,64.76 Wh,209.34 Vrms,60.02 Hz,0,0,0
93,93,8/4/2019 12:39,12538318,12538318,16S,ABC,50.00 A,21.6,quick,Optical Pickup,40454611,100.24%,100.24%,100.24%,100.22%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,19 pulses,2 pulses,10 pulses,4,50.00 A,5.00 A,50.00 A,409.43 Wh,43.10 Wh,215.52 Wh,483.45 Vrms,60.00 Hz,0,0,0
92,92,8/4/2019 12:11,12538318,12538318,16S,ABC,30.00 A,21.6,quick,Optical Pickup,40439445,100.01%,100.03%,100.01%,100.00%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,5 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,107.99 Wh,21.59 Wh,64.80 Wh,212.20 Vrms,60.01 Hz,0,0,0
91,91,8/4/2019 11:43,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40474489,100.06%,100.07%,100.06%,99.98%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.19 Wh,21.60 Wh,245.59 Vrms,59.98 Hz,0,0,0
90,90,8/4/2019 11:26,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40474562,100.11%,100.10%,100.10%,100.05%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.15 Wh,7.19 Wh,21.59 Wh,244.12 Vrms,60.00 Hz,0,0,0
89,89,8/4/2019 11:07,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40475660,100.04%,100.04%,100.04%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.18 Wh,7.20 Wh,21.59 Wh,243.99 Vrms,59.98 Hz,0,0,0
88,88,8/4/2019 10:48,12538318,12538318,2S,AC,50.00 A,12,quick,Optical Pickup,40494232,99.96%,99.95%,99.96%,99.95%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,50.00 A,5.00 A,50.00 A,72.03 Wh,12.01 Wh,36.02 Wh,243.20 Vrms,59.97 Hz,0,0,0
87,87,8/4/2019 10:22,12538318,12538318,2S,AC,30.00 A,7.2,quick,Optical Pickup,40476493,100.07%,100.10%,100.07%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,30.00 A,3.00 A,30.00 A,43.17 Wh,7.19 Wh,21.60 Wh,244.40 Vrms,59.99 Hz,0,0,0
33,33,26/06/2019 11:55,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,582792601,99.85%,99.87%,99.85%,99.82%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.06 Wh,4.01 Wh,20.04 Wh,237.94 Vrms,60.02 Hz,33.76761,-78.77817,2.52
32,32,26/06/2019 09:41,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,G009669125,99.84%,99.87%,99.84%,99.80%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.07 Wh,4.01 Wh,20.04 Wh,239.99 Vrms,60.02 Hz,33.76764,-78.77801,4.24
31,31,25/06/2019 15:16,12711729,12711729,2S,AC,30.00 A,1,custom,Optical Pickup,G009669125,-,99.92%,N/A,-,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,-,4 pulses,-,4,-,3.00 A,-,-,4.00 Wh,-,237.24 Vrms,59.96 Hz,33.76771,-78.77812,2.89
30,30,25/06/2019 15:14,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,G009669125,99.90%,99.91%,99.90%,99.88%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.04 Wh,4.00 Wh,20.02 Wh,237.32 Vrms,59.96 Hz,33.76771,-78.77812,2.89
28,28,25/06/2019 14:08,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,38942.21319,99.84%,99.87%,99.85%,99.82%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.06 Wh,4.01 Wh,20.04 Wh,238.76 Vrms,60.01 Hz,33.76759,-78.77812,4.23
27,27,25/06/2019 14:00,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,963646,99.81%,99.86%,99.82%,99.81%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.07 Wh,4.01 Wh,20.04 Wh,238.29 Vrms,60.00 Hz,33.76752,-78.77821,4
26,26,25/06/2019 12:53,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.98%,99.99%,99.99%,100.01%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,88 pulses,9 pulses,44 pulses,4,5.00 A,0.50 A,5.00 A,26.40 Wh,2.70 Wh,13.20 Wh,236.61 Vrms,60.02 Hz,0,0,0
25,25,25/06/2019 12:48,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,100.00%,99.98%,100.00%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,88 pulses,9 pulses,44 pulses,4,5.00 A,0.50 A,5.00 A,26.40 Wh,2.70 Wh,13.20 Wh,237.25 Vrms,60.01 Hz,0,0,0
24,24,25/06/2019 12:35,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.99%,99.98%,99.99%,100.04%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,88 pulses,9 pulses,44 pulses,4,5.00 A,0.50 A,5.00 A,26.40 Wh,2.70 Wh,13.19 Wh,237.56 Vrms,59.98 Hz,33.76756,-78.7782,2.44
23,23,25/06/2019 12:05,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,100.00%,99.99%,100.00%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,88 pulses,9 pulses,44 pulses,4,5.00 A,0.50 A,5.00 A,26.40 Wh,2.70 Wh,13.20 Wh,235.99 Vrms,60.01 Hz,33.76756,-78.77818,3.71
22,22,25/06/2019 10:32,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,100.01%,100.01%,100.01%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,89 pulses,9 pulses,45 pulses,4,5.00 A,0.50 A,5.00 A,26.70 Wh,2.70 Wh,13.50 Wh,238.77 Vrms,60.02 Hz,33.76755,-78.77814,2.38
21,21,25/06/2019 09:21,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.98%,99.99%,99.98%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,88 pulses,9 pulses,44 pulses,4,5.00 A,0.50 A,5.00 A,26.41 Wh,2.70 Wh,13.20 Wh,237.57 Vrms,60.01 Hz,33.76762,-78.7781,2.45
20,20,25/06/2019 07:52,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.99%,100.01%,99.99%,100.04%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,89 pulses,9 pulses,45 pulses,4,5.00 A,0.50 A,5.00 A,26.70 Wh,2.70 Wh,13.49 Wh,240.07 Vrms,60.01 Hz,33.7676,-78.77809,3.85
19,19,25/06/2019 07:22,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,1882727,100.00%,99.98%,99.99%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,90 pulses,9 pulses,45 pulses,4,5.00 A,0.50 A,5.00 A,27.00 Wh,2.70 Wh,13.50 Wh,240.80 Vrms,60.02 Hz,33.76764,-78.77818,2.97
18,18,24/06/2019 16:46,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.99%,99.98%,99.99%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,86 pulses,9 pulses,43 pulses,4,5.00 A,0.50 A,5.00 A,25.80 Wh,2.70 Wh,12.90 Wh,230.06 Vrms,59.99 Hz,33.76759,-78.77819,2.69
16,16,24/06/2019 15:40,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.99%,100.00%,99.99%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,86 pulses,9 pulses,43 pulses,4,5.00 A,0.50 A,5.00 A,25.80 Wh,2.70 Wh,12.90 Wh,230.61 Vrms,60.02 Hz,33.7676,-78.77809,2.2
15,15,24/06/2019 14:40,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,100.00%,99.98%,99.99%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,88 pulses,9 pulses,45 pulses,4,5.00 A,0.50 A,5.00 A,26.40 Wh,2.70 Wh,13.50 Wh,237.26 Vrms,59.99 Hz,33.76757,-78.77809,2.61
14,14,24/06/2019 14:23,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,100.00%,100.02%,100.00%,100.04%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,89 pulses,9 pulses,45 pulses,4,5.00 A,0.50 A,5.00 A,26.70 Wh,2.70 Wh,13.50 Wh,237.68 Vrms,60.00 Hz,33.76753,-78.77821,2.44
13,13,24/06/2019 14:08,12711729,12711729,6FS,ABC,5.00 A,0.3,quick,Optical Pickup,582792601,99.99%,99.98%,99.99%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,89 pulses,9 pulses,45 pulses,4,5.00 A,0.50 A,5.00 A,26.70 Wh,2.70 Wh,13.50 Wh,237.67 Vrms,59.99 Hz,33.76763,-78.77806,2.01
12,12,24/06/2019 13:05,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,582792601,99.87%,99.91%,99.88%,99.89%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.05 Wh,4.00 Wh,20.02 Wh,236.44 Vrms,60.02 Hz,33.76758,-78.77813,2.38
11,11,24/06/2019 12:51,12711729,12711729,2S,AC,30.00 A,1,quick,Optical Pickup,582792601,99.84%,99.87%,99.84%,99.83%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,40 pulses,4 pulses,20 pulses,4,30.00 A,3.00 A,30.00 A,40.07 Wh,4.01 Wh,20.03 Wh,235.54 Vrms,60.01 Hz,33.76763,-78.77817,2.62
10,10,24/06/2019 09:57,12711729,12711729,4S,AC,2.50 A,0.6,quick,Optical Pickup,FOCUS4S,99.89%,99.91%,99.89%,99.97%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,2.50 A,0.25 A,2.50 A,3.60 Wh,0.60 Wh,1.80 Wh,236.87 Vrms,60.01 Hz,0,0,0
9,9,24/06/2019 09:36,12711729,12711729,4S,AC,2.50 A,0.6,quick,Optical Pickup,101388118,99.86%,99.89%,99.87%,99.95%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,6 pulses,1 pulses,3 pulses,4,2.50 A,0.25 A,2.50 A,3.61 Wh,0.60 Wh,1.80 Wh,238.75 Vrms,59.98 Hz,33.76755,-78.77821,2.42
8,8,12/6/2019 18:05,12711729,12711729,16S,ABC,30.00 A,21.6,quick,Optical Pickup,G9SR0013649,100.07%,100.05%,100.06%,100.06%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,9 pulses,1 pulses,5 pulses,4,30.00 A,3.00 A,30.00 A,194.27 Wh,21.59 Wh,107.93 Wh,356.30 Vrms,60.02 Hz,46.46655,-71.02393,199.47
7,7,22/03/2019 16:28,12711729,12711729,16S,ABC,30.00 A,21.6,quick,Optical Pickup,TEDPX000673825,100.06%,100.07%,100.06%,100.03%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,3 pulses,1 pulses,2 pulses,4,30.00 A,3.00 A,30.00 A,64.76 Wh,21.58 Wh,43.19 Wh,117.30 Vrms,59.98 Hz,21.30068,-157.85387,16.45
5,5,22/03/2019 15:23,12711729,12711729,9S,ABC,2.50 A,1.8,quick,Optical Pickup,KZFPX000648310,99.98%,99.96%,99.96%,99.92%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,3 pulses,1 pulses,2 pulses,4,2.50 A,0.25 A,2.50 A,5.40 Wh,1.80 Wh,3.60 Wh,117.11 Vrms,59.98 Hz,0,0,0
4,4,20/03/2019 21:03,12711729,12711729,9S,ABC,2.50 A,1.8,quick,Optical Pickup,PX000563716,99.97%,99.97%,99.96%,99.92%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,3 pulses,1 pulses,2 pulses,4,2.50 A,0.25 A,2.50 A,5.40 Wh,1.80 Wh,3.60 Wh,119.53 Vrms,60.00 Hz,0,0,0
2,2,30/12/2012 19:20,12711729,12711729,9S,ABC,2.50 A,1.8,quick,Optical Pickup,MPX000653953,100.04%,100.01%,100.03%,100.02%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,3 pulses,1 pulses,2 pulses,4,2.50 A,0.25 A,2.50 A,5.40 Wh,1.80 Wh,3.60 Wh,118.09 Vrms,60.03 Hz,21.30077,-157.85393,16.1
6,6,30/12/2012 19:08,12711729,12711729,16S,ABC,30.00 A,21.6,quick,Optical Pickup,TEDPX000673825,100.03%,100.04%,100.03%,100.00%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,3 pulses,1 pulses,2 pulses,4,30.00 A,3.00 A,30.00 A,64.78 Wh,21.59 Wh,43.20 Wh,117.28 Vrms,59.98 Hz,21.30068,-157.85387,16.45
3,3,30/12/2012 19:04,12711729,12711729,9S,ABC,2.50 A,1.8,quick,Optical Pickup,KZFPX000563895,99.98%,99.98%,99.97%,99.96%,N/S,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,N/A,3 pulses,1 pulses,2 pulses,4,2.50 A,0.25 A,2.50 A,5.40 Wh,1.80 Wh,3.60 Wh,117.91 Vrms,60.01 Hz,21.30068,-157.85396,15.5